OTSC® application aid for better approximation of tissue especially when indurated .
Use of the OTSC® Anchor
The OTSC® Anchor can be used for better approximation of tissue especially when indurated (e.g. fistulae, chronic ulcer). It also facilitates targeting of lesions, e.g. in the treatment of hemorrhage, the OTSC® Anchor allows precise alignment between the target tissue and the applicator cap.
Dimensions and specifications
The OTSC® Anchor is available in two versions and lengths:
- The OTSC® Anchor has a flexible shaft length of 165 cm
- Der OTSC® The OTSC® Anchor 220 tt is 220 cm in length and especially for thin tissue
|OTSC® Anchor||OTSC® Anchor 220tt|
|Needle width||12 mm||9 mm|
|Stich depth||4 mm||2 – 2.5 mm|
|Length||165 cm||220 cm|
|Appropriate endoscope||working channel diameter of minimum 2.8 mm to be used alone or 3.2 mm with the OTSC® System Set.||working channel diameter of minimum 2.8 mm to be used alone or 3.2 mm with the OTSC® System Set.|
|Items per package||1||1|
Clinical procedure: Use of OTSC® Anchor as a guiding instrument
OTSC® Anchor positioning and tissue fixation; align the OTSC® cap to the lesion by pulling the Anchor and advancing the endoscope
Mobilize the tip of the OTSC® Anchor shaft into cap. Anchor spikes may remain external. Clip release
After clip application detach the OTSC® Anchor from the tissue.
Closure of persistent PEG-fistula using the OTSC® Anchor
*Source: Dr. Thomas Kratt, Interventional Endoscopy, Klinik für Allgemeine, Viszeral- und Transplantationschirurgie, University Hospital Tuebingen, Germany
Example of application: Optimized fistula closure
Mucosal incision (circular; not too deep, because of perforation risk) around the fistula opening with the AqaNife (Ø ~ 15 mm)
Positioning of the clip in the incision line
Clip application and closure of the fistula
*Source: Meining A. et al. (2015) Erfolgreicher Verschluss einer großen ösophago-bronchialen Fistel durch mukosale Inzision vor OTSC-Clip-Platzierung. Endoskopie heute. Doi: 10.1055/s-0035-1545049